Baxter 76301 - PUMP, INFUSION, BAXTER, FLOGARD 6301, DUAL, EACH
Flo-Gard 6301, Dual Channel Volumetric Infusion Pump
The Flo-Gard 6301 Dual Channel Volumetric Infusion Pump is an electromechanical device used for the intravenous infusion of liquids at user-selected rates. The device contains two linear peristaltic pump heads, allowing it to simultaneously infuse two different solutions. Each pump head is independently programmable and permits infusion of primary and secondary medication programs. The secondary program automatically switches over to the primary program when secondary infusion is complete (automatic piggybacking).
The device operates on standard 115 VAC 60 Hz electrical power, or on its self-contained rechargeable battery. It is portable and has a panel lock-out feature to prevent patient tampering. It is designed for use with Baxters standard administration sets which contain an s as the last character of the code number, for example, 2C5537s. When infusing solutions through a central line catheter, sets with Luer lock adapter should be used. Sets with a Flashball@ device are not recommended in these applications.
The primary rate of infusion is selectable from 1 to 1,999 mL in 1 mL/hr incrementsand 1 .O to 99.9 mL/hr in 0.1 mL/hr increments. The secondary rate is selectable from 1 to 999 mL in 1 mL/hr increments and 1 .O to 99.9 mL/hr in 0.1 mL/hr increments. The volume to be infused (VTBI) is also selectable from 1 to 9,999 mL in 1 mL increments and 1 .O to 99.9 mL in 0.1 mL increments.
The total volumes infused from primary and secondary programs are added together and accumulated and can be displayed on demand. The primary and secondary VTBIs are independently decremented and displayed. Upon completion of the primary VTBI, the pump automatically switches to a keep vein open (KVO) rate. If the pump is started on a secondary rate and VTBI, the pump will change to the primary rate when the secondary program is completed. Either pump may be stopped at any time by depressing the STOP key unless the device is in the lockout mode.
|Description||Dual channel linear peristaltic volumetric infusion pump|
|Administration Set||Baxters standard administration set with "s" suffix|
|Keep Vein Open (KVO) rate||5 mL/hr or programmed rate, whichever is less|
|Battery||12 Volt, 3.2 Ah sealed lead acid|
|Battery Recharge||8 hours for 80% recharge|
|AC Power Requirements||11 O/f20V, 60 Hz|
|Power Cord||2.75 m (9.0 ft.) long, with Hospital Grade plug|
|Leakage Current||Typically less than 50 microamps (using UL-544 specified test methods)|
|Power Consumption||50 watts|
|Weight||8.2 kg (18 Ibs)|
|Flow Rate Range|
|VT81 Range||I - 9,999 mL in 1 mL increments and 1 .O - 99.9 in 0.1 mL increments for both primary and secondary. Upper limit can be reduced by authorized service personnel.|
|Air-in-Line Detection||Factory set to NORM which causes the device to alarm on air bubbles approximately 75pL or larger. The MIN setting causes the device to alarm on bubbles approximately 50 PL or larger.|
|Fuse||0.5 A Slo-610|
|Nurse Call||g-pin D connector, Pin 1: N/C (normally closed), Pin 4: N/O (normally open), Pin 9: Common Contact rating: 0.4A at 30 VAC, 2A at 30 VDC resistive load (internal connection required)|