C.R. Bard 136505 - Polyurethane Ureteral Catheters, 70 cm, 5 FR, Olive Tip, with Adapter, 10/CS
C.R. Bard 136505 Polyurethane Ureteral Catheters
BD manufactures high quality single-use polyurethane ureteral catheters that provide excellent handling characteristics, proper flexibility and resistance to kinking. Ureteral catheters are available in a variety of sizes and tip configurations. They come packaged sterile for single use.
- Flexible Tip Designed to allow access to a tortuous ureter.
- Shaft Quality BD Polyurethane Ureteral Catheters have an exclusive high quality shaft that is flexible and smooth for greater patient comfort.
- Ease of Identification Color-coded distal end for ureteral identification; Red-right, Blue left.
- Ease of UseSingle use product for convenience and assured sterility.
- Complimentary Line Various sizes and tip configurations for a variety of procedures.
- All polyurethane ureteral catheters are radiopaque, graduated and packaged sterile.
|136505||Olive Tip||70 cm, 5 FR|
C.R. Bard 136505 Polyurethane Ureteral Catheters Indications for Use:
The Bard Polyurethane ureteral catheters are intended for use in the drainage of urine from the ureter. The polyurethane ureteral catheter is inserted through the urethra, into the bladder and up through the ureter. These catheters also allow the delivery of contrast media.
Directions for Use of C.R. Bard 136505 Polyurethane Ureteral Catheter
Ureteral catheters are designed in a variety of distal tip configurations. The anatomical characteristics of the ureteral orifice and physician preference should determine the choice of design in a specific clinical situation.
Closed Tip Placement (placed without a guidewire)
After placement of the cystoscope into the bladder, the ureteral orifice is located and examined and a choice of ureteral catheter is made. The catheter adapter is secured to the ureteral catheter, and the catheter is flushed to evacuate all air bubbles. Under direct vision, the catheter tip is positioned within the ureteral orifice. Contrast media is then injected through the ureteral catheter into the ureter. Under fluoroscopy, the ureter is visualized and required intervention determined.
Open Tip Placement (placed with a guidewire)
After placement of the cystoscope into the bladder, the ureteral orifice is located and examined and a choice of ureteral catheter is made. Under direct vision, the guidewire is positioned within the ureteral orifice. The ureteral catheter is inserted over the guidewire and advanced into the ureter.
The guidewire is removed and the catheter adapter is secured to the catheter. Gentle aspiration to evacuate air from the catheter can be performed at this time. Contrast media is then injected through the ureteral catheter into the ureter. Under fluoroscopy, the ureter is visualized and required intervention determined.
C.R. Bard 136505 Polyurethane Ureteral Catheters Device Characteristics
|What MRI safety information does the labeling contain?||Labeling does not contain MRI Safety Information|
|Device required to be labeled as containing natural rubber latex or dry natural rubber (21 CFR 801.437):||No|
|Device labeled as "Not made with natural rubber latex":||No|
|Prescription Use (Rx):||Yes|
|Over the Counter (OTC):||No|
|Human Cell, Tissue or Cellular or Tissue-Based Product (HCT/P):||No|